Welcome to the website of Oriol Arteaga. Here you will mostly hear about my research activities which mainly focus in polarimetry, ellipsometry or, more in general, in light-matter interactions. You can contact me at oarteaga(AT)ub.edu
For over a year and a half I have been working on Mueller matrix microscopy. Now I can say that I have no doubts that, when it is properly implemented, Mueller matrix microscopy is nothing but the perfect upgrade for polarization microscopy. I know polarization microscopy is not the most trendy techniquein these days, but polarization microscopes are so widespread that even if only a tiny fraction of core polarization microscopy users become aware of what Mueller matrix microscopy can offer, the usage and the applications for Mueller matrix microscopy will grow exponentially.
Detail of the scale of a Morpho butterfly captured with a Mueller microscope.
With this objective in mind we have been working in the last few months on a compact, table-top prototype of a Mueller matrix microscope. The skeleton and the optics of the instruments are based on commercial polarization microscope, but it has been heavily modified to work as a fully-automatized Mueller matrix microscope. The aim was to keep all the advantages of polarization microscopy, with a very compact instrument in which it is very easy to manipulate the sample and find a focus but, at the same time, all the power of Mueller matrix analysis is just available after a “click”. The working principle of this Mueller microscope was described in Oriol Arteaga, et al. “Mueller matrix microscope with a dual continuous rotating compensator setup and digital demodulation,” Appl. Opt. 53, 2236-2245 (2014).
Photo of our benchtop Mueller matrix miscroscope. It is based on classic ZEISS Jenaval microscope that has been modified.
The project is being carried with the technical collaboration of A.COLOMA Microscopios y Aparatos Ópticos and we work together to offer complete solutions for Mueller matrix microscopy. At this early stage, you can contact me (oarteagaATub.edu) or Francesc Gomez (fxgomezATacolomamicroscopis.com) in case you are interested on getting a Mueller matrix microscope or if you would like to make a custom modification on your polarization microscope.
UPDATE! Click on the following image the pdf version of a presetation about MM microscopy that I did some time ago.
It is time to welcome 2015. This New Year should be specially interesting for us since it is the International Year of Light and Light-based Technologies. The International Year of Light is a global initiative adopted by the United Nations to raise awareness of how optical technologies promote sustainable development and provide solutions to worldwide challenges in energy, education, agriculture, communications and health.
From this tiny spot I hope I can contribute with a grain of sand to these goals by showing how a property of light that we can not perceive, polarization, becomes our tool to investigate many properties of the world that surround us.
After a long time without updates, I have just renewed the design of the website. My New Year’s resolution should be to update this website much more often.
In the last few months a good part of my research has been about Mueller matrix microscopy. We have developed a new MM microscope that uses a dual rotating compensator setup and digital demodulation. The spatial resolution and measurement quality is really good. See related publications here and here.
At the beginning of May I attended SPIE.DSS conference 2014 in Baltimore (Maryland, USA). I had never been in a conference that huge. Also I took advantage that Baltimore is close to NYC to visit again my old laboratory. I was happy to see that my old 4PEM MM polarimeter is doing fine.
For some reason I cannot fully understand, certain types of serial decompositions of Mueller matrices became quite popular among experimentalists. I am talking about the polar decomposition and the Lu-Chipman decompositions. These matrix decompositions allow to decompose a matrix in a series of a retarder, a diattenuator and, eventually, also a depolarizer. Of course matrix multiplication is not commutative and, therefore, these decompositions are not unique. In general my advise would be to avoid serial decomposition of a Mueller matrix unless you have reasonable suspects that the investigated sample really corresponds to a certain succession of a retarder, diattenuator and a depolarizer. And, as a experimentalist, I can say that this will rarely happen.
In my view, the decompositions of the Mueller matrix based on a continuous medium have a much more general applicability and they should be the best choice for unknown samples. Some years ago we published the analytic equations find the results straightforwardly (Optics letters 35.4 (2010): 559-561 and Optics Letters 35.20 (2010): 3525-3525). However such equations apply to a Mueller-Jones matrix (non-depolarizing Mueller matrix) and I am aware that some people is having difficulties to apply them to general (depolarizing) Mueller matrix. In the papers that procedure was outlined, but anyway I upload here the code for such calculation, to clarify how the results should look like.
I upload a Labview library program that does such interpretation (run the vi “MManalysis.vi”. I also upload the same program but as an application in case there is anybody that cannot use Labview. Once you open it you will find something like this:
You only need to write down a Mueller matrix (preferably an experimental one) and the program will give the closest Mueller-Jones and Jones matrices (in the Cloude sense) and the 6 independent parameters that better define your media (CD,CB,LD,LB,LD’ and LB’). **
**. This analytic interpretation of continuous medium coincides 100% with the Log/differential method of analysis of Mueller matrices for nondepolarizing media, and it is computationally faster and numerically more robust. In the presence of strong depolarization there will be differences among both techniques, but that will be a topic for another day.
During 2014-2015 I will be working on a European Marie Curie IIF project, titled Polarimetric characterization of natural and artificial chiral anisotropic media (Nanochirality) at University of Barcelona.
Not many updates lately! I hope this semester I will manage to keep the website more up-to-date. I also want to add a section with some new scientific content in the form of a tutorial, let’s see if I find the time. For the moment today I have updated the publication list in the lazy way, with a link to Google Schoolar. You can check my latest publications there. Any kind of feedback will be very appreciated!
Hinds Intruments is selling the new 150XT Mueller Polarimeter. This instrument was developed in collaboration with New York University using the 4 Photoelastic Modulator (PEM) technology we pioneered. To my knowledge this is the first commercial full Mueller matrix polarimeter with research-grade specifications ever made available . The key element here is the sensitivity. The instrument was developed to solve complex optical activity measurements involving chiral molecules molecules and crystals that show the tiniest bias to circular polarization. The result is a highly versatile instrument, capable of studying the transmitted complicate optical response of virtually any sample, even highly depolarizing one. The tiniest strain in a glass, the optical rotation of a birefringent crystal, the circular dichroism of a oriented molecular aggregate, the heterogeneous response of a petrological sample, and many other can be studied with this polarimeter.
You can have more information about this instrument checking their brochure:
After more than two years in New York University this month I have started to work again in Universitat de Barcelona. This was possible because some time ago I got a Beatriu de Pinos Fellowship. These are grants to encourage the recruitment of research personnel in the Catalan public sector, be they in universities, research centers or hospital foundations, who have had postdoctoral stays abroad.
Lately I have been quite busy with the move but now I am already resettled. I pretend to keep working in the same reseach areas as before so expect here more stuff about Mueller Matrix polarimetry and others.
Recently I have been working on how Mueller matrices handle reciprocal and nonreciprocal optical effects. Mueller matrices are truly informativative about reciprocity, but one needs to know how the information is “stored”. A series of two papers are going to be published in Optics Letters about the topic. For the moment they can be accessed in early view.